One of the instruments I have is a HP 53131A Frequency Counter, and putting the positives of this unit aside, its by far the most annoying bit of test gear I have! Why? well for some reason, when you apply power to the thing the fan runs even when its switched off. This is because the power switch is a soft switch and with just the AC cord plugged in the fan runs and makes the annoying fan noise. This is a really poor design and certainly not one of HP’s shining examples of engineering….I like to be able to switch my stuff off without reaching around the back and having to pull the power chord or unplug from the wall outlet – just plain crappy! HP what where you thinking?
I thought I would tear it down and have a look and see why this is, and perhaps see if there is any possible modification I could come up with to improve on this. When I took it apart I found a real mess of dust and some kind of substance spillage so a full clean-up was needed which somewhat distracted me from the power switch problem.
While its in bits, I also do a quick run through of the major components of the circuit and have a quick look at the power supply and power switch circuits in some detail.
UPDATE: The clean-up also appears to have resolved the strange trigger problems I had seen previously with this counter. Well pleased…
Here is the component level manual including schematics for the frequency counter
Thanks for watching…
Tibs are pocket-change for the internet™.