It all started when I wanted to calibrate my HP 53131A universal counter, which as it turns out probably has one of the crappiest and most disappointing standard oscillators ever put into a frequency counter, HP you should bow your head in shame….oh of course I forgot, a half reasonable oscillator is an “optional extra” when you by HP/Agilent – of course it is….anyway, on with the job at hand
If you have or want to play with an FE-5680A Rubidium Frequency Standard or an OSCILLOQUARTZ OCXO 8663-XS or a HP 53131A Counter or a Racal Dana 1999 counter or similar then this video will most likely be of interest 🙂 what I am trying to get is a predicable and reliable frequency and standard for my home lab.
I guess I will let the video do the talking on this one….
Here are a whole bunch of useful links that relate to this video (there are many more too if you search around the web)
- FE-5680A Manufacturer Page
- The (very good) e-bay seller I mention in the video
- KO4BB"s Wiki
- A good document about the internals and programming a FE-5680A by Clint (KA7OEI)
- Another good source of information about programming a FE-5680A by Murray Greenman (ZL1BPU)
- A good FAQ about the internals and programming a FE-5680A by John Beale
- Some more info on the FE-5680A
- A teardown of an FE-5680A by Dave Jones EEVBlog
- An even deeper teardown of an FE-5680A by Mike @ MikesElectricStuff
- A simple interface board for an FE-5680A
- Stability and Performance Notes by John Miles, KE5FX
- |Information on some modifications of an FE-5680A by MMatthias
- Datasheet for OSCILLIQUARTZ OCXO 8663-XS
- Ian Johnston’s Racal Dana 1991 Frequency Counter Repair
Thanks to all of the authors and content creators for the above information. Thanks for watching.
This content is published under the Attribution-Noncommercial-Share Alike 3.0 Unported license.
Tibs are pocket-change for the internet™.