I have had my HP 339A for a long time now and despite only using it a small handful of times it is a really great instrument – but – its big and takes up a lot of shelf space. A few weeks back a purchased a Keithley 2015 THD which is a 6.5 digit bench meter with a built in distortion analyser, I bought it because of the distortion analyser function. I would like to get some more instrument shelf space back so am able to replace the HP 339 with this 2015THD I will make that much needed space. I decided to build a simple output circuit so that I can induce some crossover distortion and compare the basic measurements on these two devices to see the results. Now I must make it clear that this is a basic 101 on measuring distortion, I am sure there is a lot more to know that I do but I believe I cover off the basis.
I will tear down the HP 339A on video to show how nicely these are built. This is one of those rare “all analog” devices that has not digital parts at all, watch out for that video if you like seeing the guts of nicely built equipment.
Thanks for watching.
This content is published under the Attribution-Noncommercial-Share Alike 3.0 Unported license.
Tibs are pocket-change for the internet™.